The seems to put out negative voltage spikes in the range of -5V to if the current is switched to mA there is a large negative spike on.
All Keithley product names are trademarks or registered trademarks of Keithley formation, and as shown on the instrument or test fixture panels, or switching card . During power-up, voltage spikes may appear on the terminals of the.
Is your voltage reading set to autorange? If not, then you're probably reaching range-compliance. Throw in a Keithley "Configure.
Trigger-Link interface to Keithley Series switching hardware. . Power-up. During the power-up, voltage spikes may appear on the terminals of.
Keithley SourceMeter review and teardown - Page 1. Under analog board we can find switching power supply and digital brains board. Thanks to SMPS .. This is because of a spike on the 15 and rail at powerup.
During the power-up, voltage spikes may appear on the terminals of the SourceMeter. These voltage 1) No DUT connected while powering up (turn ON sequence of the instrument) 2. Return to “ Series SourceMeter”.